BEACON eSpace at Jet Propulsion Laboratory >
JPL Technical Report Server >
JPL TRS 1992+ >
Please use this identifier to cite or link to this item:
|Title: ||A first order wavefront estimation algorithm for P1640 calibrator|
|Authors: ||Zhaia, C.|
Roberts, L. C. Jr.
|Keywords: ||Wavefront sensing|
high contrast imaging
|Issue Date: ||1-Jul-2012 |
|Publisher: ||Pasadena, CA : Jet Propulsion Laboratory, National Aeronautics and Space Administration, 2012.|
|Citation: ||2012 Astronomical Telescopes + Instrumentation, Amsterdam, Netherlands, July 1-6, 2012|
|Abstract: ||P1640 calibrator is a wavefront sensor working with the P1640 coronagraph and the Palomar 3000 actuator adaptive optics system (P3K) at the Palomar 200 inch Hale telescope. It measures the wavefront by interfering post-coronagraph light with a reference beam formed by low-pass filtering the blocked light from the coronagraph focal plane mask. The P1640 instrument has a similar architecture to the Gemini Planet Imager (GPI) and its performance is currently limited by the quasi-static speckles due to non-common path wavefront errors, which comes from the non-common path for the light to arrive at the AO wavefront sensor and the coronagraph mask. By measuring the wavefront after the coronagraph mask, the non-common path wavefront error can be estimated and corrected by feeding back the error signal to the deformable mirror (DM) of the P3K AO system. Here, we present a first order wavefront estimation algorithm and an instrument calibration scheme used in experiments done recently at Palomar observatory. We calibrate the P1640 calibrator by measuring its responses to poking DM actuators with a sparse checkerboard pattern at different amplitudes. The calibration yields a complex normalization factor for wavefront estimation and establishes the registration of the DM actuators at the pupil camera of the P1640 calibrator, necessary for wavefront correction. Improvement of imaging quality after feeding back the wavefront correction to the AO system demonstrated the efficacy of the algorithm.|
|Appears in Collections:||JPL TRS 1992+|
Items in DSpace are protected by copyright, but are furnished with U.S. government purpose use rights.