BEACON eSpace at Jet Propulsion Laboratory >
JPL Technical Report Server >
JPL TRS 1992+ >
Please use this identifier to cite or link to this item:
|Title: ||Development of fast, background-limited transition-edge sensors for the Background-Limited Infrared/Sub-mm Spectrograph (BLISS) for SPICA|
|Authors: ||Beyer, Andrew D.|
Runyan, M. C.
Echternach, P. M.
Bradford, C. M.
Holmes, W. A.
Bock, J. J.
|Keywords: ||transition-edge sensors|
|Issue Date: ||2-Jul-2012 |
|Publisher: ||Pasadena, CA : Jet Propulsion Laboratory, National Aeronautics and Space Administration, 2012.|
|Citation: ||SPIE Astronomical Telescopes and Instrumentation 2012, Amsterdam, Netherlands, July 1-6, 2012|
|Abstract: ||We report experimental progress toward demonstrating background-limited arrays of membrane-isolated transition-edge sensors (TESs) for the Background Limited Infrared/Sub-mm Spectrograph (BLISS). BLISS is a space-borne instrument with grating spectrometers for wavelengths λ= 35-435μm and with R=λ/Δλ~500. The goals for BLISS TESs are: noise equivalent power (NEP) = 5×10-20 W/Hz1/2 and response time τ<30ms. We expect background-limited performance from bilayers TESs with Tϲ=65mK and G=15fW/K. However, such TESs cannot be operated at 50mK unless stray power on the devices, or dark power PD, is less than 200aW. We describe criteria for measuring PƉ that requires accurate knowledge of TC. Ultimately, we fabricated superconducting thermistors from Ir (Tϲ≥135mK) and Mo/Cu proximitized bilayers, where Tϲ is the thermistor transition temperature. We measured the Ir TES arrays in our 50mK adiabatic demagnetization refrigerator test system, which can measure up to eight 1x32 arrays simultaneously using a time-division multiplexer, as well as our single-pixel test system which can measure down to 15mK. In our previous Ir array measurements our best reported performance was NEP=2.5×10-19 W/Hz1/2 and ʈ~5ms for straight-beam TESs. In fact, we expected NEP~1.5×10-19W/Hz1/2 for meander beam TESs, but did not achieve this previously due to 1/f noise. Here, we detail improvements toward measuring the expected NEP and demonstrate NEP=(1.3+0.2)×10- 19W/Hz1/2 in our single-pixel test system and NEP=(1.6+0.3)×10-19W/Hz1/2 in our array test system.|
|Appears in Collections:||JPL TRS 1992+|
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.